Heat Stress Monitor by Using Low-Cost Temperature and Humidity Sensors
The aim of this study is to develop a cost-effective WBGT heat stress monitor which provides precise heat stress measurement. The proposed device employs SHT15 and DS18B20 as a temperature and humidity sensors, respectively, incorporating with ATmega328 microcontroller. The developed heat stress monitor was calibrated and adjusted to that of the standard temperature and humidity sensors in the laboratory. The results of this study illustrated that the mean percentage error and the standard deviation from the measurement of the globe temperature was 2.33 and 2.71 respectively, while 0.94 and 1.02 were those of the dry bulb temperature, 0.79 and 0.48 were of the wet bulb temperature, and 4.46 and 1.60 were of the relative humidity sensor. This device is relatively low-cost and the measurement error is acceptable.
Digital Object Identifier (DOI): doi.org/10.5281/zenodo.1073317Procedia APA BibTeX Chicago EndNote Harvard JSON MLA RIS XML ISO 690 PDF Downloads 2082
 D.S. Moran et al. An environmental stress index (ESI) as a substitute for the wet bulb globe temperature (WBGT). Thermal Biology 26 (2001) 427-431
 Jose Perez-Alonso et al. Approach to the evaluation of the thermal work environment in the greenhouse-construction industry of SE Spain. Building and Environment 46 (2011) 1725e1734
 American Conference of Governmental Industrial Hygienists. TLV & BEI based on the documentation for the threshold limit values for chemical substance and physical agents & biological exposure indices. 2008
 Chalermchai Chaikittiporn. Safety and environment for industrial. Department of Environmental Quality Promotion, Ministry of Natural Resources and Environment
 Pornpimol Kongtip. Industrial hygiene, Department of occupational health and safety, Faculty of public health, Mahidol University.
 Wantanee Phanprasit, Mahidol University. The guideline for heat, light and sound monitoring B.E. 2549. 2006
 Thailand Regulations, set standards for occupational health, safety and environment to work on heat, light and sound B.E. 2549. 2006
 Aekkachai Makarn. Learn to use the AVR family of microcontrollers with Arduino, ETT Company Limited
 Occupational safety and health bureau, Department of labour protection and welfare, Sampling and Evaluation of Physical Environments Manual, 2002
 G. Buonanno, A. Frattolillo, L. Vanoli. Direct and indirect measurement of WBGT index in transversal flow, Measurement 29 (2001) 127-135
 Grahame M. Budd. Wet-bulb globe temperature (WBGT) - its history and its limitations, Science and Medicine in Sport (2008) 11, 20 - 32