Commenced in January 2007
Paper Count: 30075
Dosimetric Comparison of aSi1000 EPID and ImatriXX 2-D Array System for Volumetric Modulated Arc and Intensity Modulated Radiotherapy Patient Specific Quality Assurance
Abstract:Prior to the use of detectors, characteristics comparison study was performed and baseline established. In patient specific QA, the portal dosimetry mean values of area gamma, average gamma and maximum gamma were 1.02, 0.31 and 1.31 with standard deviation of 0.33, 0.03 and 0.14 for IMRT and the corresponding values were 1.58, 0.48 and 1.73 with standard deviation of 0.31, 0.06 and 0.66 for VMAT. With ImatriXX 2-D array system, on an average 99.35% of the pixels passed the criteria of 3%-3 mm gamma with standard deviation of 0.24 for dynamic IMRT. For VMAT, the average value was 98.16% with a standard deviation of 0.86. The results showed that both the systems can be used in patient specific QA measurements for IMRT and VMAT. The values obtained with the portal dosimetry system were found to be relatively more consistent compared to those obtained with ImatriXX 2-D array system.
Digital Object Identifier (DOI): doi.org/10.5281/zenodo.1061250Procedia APA BibTeX Chicago EndNote Harvard JSON MLA RIS XML ISO 690 PDF Downloads 2181
 T.Wiezorek., M. B. Nschwedas, D. Georg and T. G. Wendt, "Dosimetric quality assurance for intensity-modulated radiotherapy feasibility study for a filmless approach." Strahlenther. Oncol., vol. 81, pp. 468-74, 2005.
 B F Branci, L Marrazzo and S Russo, "An inter comparison between film dosimetry and diode matrix for IMRT quality assurance." Med. Phys., vol.34, pp. 1372-1379, 2007.
 G J Budgell, Q Zhang, R J Trouncer and R I Mackay, "Improving IMRT quality control efficiency using an amorphous silicon electronic portal imager," Med. Phys., vol.32, pp.3267-78, 2005.
 P Winkler, A Hefner and D Georg, "Dose- response characteristics of amorphous silicon EPID," Med. Phy.,; vol. 32, pp. 3095-105, 2005.
 E E Grein, R Lee and K Luchka "An investigation of a new amorphous silicon electronic portal imaging device for transit dosimetry," Med. Phys.vol 29: pp. 2262, 2002. R. W. Lucky, "Automatic equalization for digital communication," Bell Syst. Tech. J., vol. 44, no. 4, pp. 547-588, Apr. 1965.
 P. B. Greer, C. C and Popescu, "Dosimetric properties of an amorphous silicon electronic portal imaging device for verification of dynamic intensity modulated radiation therapy," Med. Phys., vol. 30, pp. 1618-27, 2003.
 E Spezi, A L Angelini, F Romani and A Ferri, "Characterization of a 2D ion chamber array for the verification of radiotherapy treatments," Phys. Med. Biol., vol. 50, pp. 3361-3373, 2005.
 B V Elsevier, "Evaluation of a 2D diode array for IMRT quality assurance," Radiother. Oncol., vol. 70, pp.199-206, 2009.
 N Agazaryan,D S Timothy and J J Demarco, "Patient specific quality assurance for the delivery of intensity modulated radiotherapy," J. Appl. Clin. Med. Phys., vol. 4, 2003.
 C D Wagter, "The ideal dosimeter for intensity modulated radiation therapy (IMRT): what is required?" Jr. Phys., vol. 3, pp. 4-8, 2004.
 W. V. Elmpt, L. Mcdermott, S. Nijsten, M. wending, et al, "A literature review of electronic portal imaging for radiotherapy dosimetry," Radiother. oncol., vol. 88, pp. 289-309, 2008.