Commenced in January 2007
Paper Count: 30998
Distortion of Flow Measurement and Cavitation Occurs Due to Orifice Inlet Velocity Profiles
Abstract:This analysis investigates the distortion of flow measurement and the increase of cavitation along orifice flowmeter. The analysis using the numerical method (CFD) validated the distortion of flow measurement through the inlet velocity profile considering the convergence and grid dependency. Realizable k-e model was selected and y+ was about 50 in this numerical analysis. This analysis also estimated the vulnerability of cavitation effect due to inlet velocity profile. The investigation concludes that inclined inlet velocity profile could vary the pressure which was measured at pressure tab near pipe wall and it led to distort the pressure values ranged from -3.8% to 5.3% near the orifice plate and to make the increase of cavitation. The investigation recommends that the fully developed inlet velocity flow is beneficial to accurate flow measurement in orifice flowmeter.
Digital Object Identifier (DOI): doi.org/10.5281/zenodo.1061096Procedia APA BibTeX Chicago EndNote Harvard JSON MLA RIS XML ISO 690 PDF Downloads 1937
 http://en.wikipedia.org/wiki/Orifice_plate, Wikipedia, 2011.
 General Requirements for Fluid Metering: Installation. USA: ASME PTC 19.5-72, 1972, pp. 179-181.
 Flow Conditioning and Meter Installation Requirements. USA: ASME PTC 19.5-2004, 2004, pp. 64-66.
 Salim et al., "Wall y+ Stratagy for Dealing with Wall-bounded Turbulent Flows", IMECS 2009, Vol II, Hong Kong, 2009
 M. Ishii, K. Mishima, Two-fluid model and hydraulic constitutive relations, Nucl. Eng. Des., Vol. 82, 1984, pp. 107.
 CFX Manual, Theory, 2009