Surface Topography Measurement by Confocal Spectral Interferometry
Confocal spectral interferometry (CSI) is an innovative optical method for determining microtopography of surfaces and thickness of transparent layers, based on the combination of two optical principles: confocal imaging, and spectral interferometry. Confocal optical system images at each instant a single point of the sample. The whole surface is reconstructed by plan scanning. The interference signal generated by mixing two white-light beams is analyzed using a spectrometer. In this work, five ‘rugotests’ of known standard roughnesses are investigated. The topography is then measured and illustrated, and the equivalent roughness is determined and compared with the standard values.
Digital Object Identifier (DOI): doi.org/10.5281/zenodo.1340304Procedia APA BibTeX Chicago EndNote Harvard JSON MLA RIS XML ISO 690 PDF Downloads 585
 Y. Usson & C. Souchier, “La tomographie en Microscopie”, Hermes Sciences, 2002, Ch. 5.
 B. Rivett et al., “Surface topography evolution through production of aluminium offset lithographic plates”, Wear, Vol. 270, Issues 3–4, 12 January 2011, pp. 204-217
 B. Michelt, J. Schulze, “The spectral colours of nanometers”, Reprint from the journal Mikroproduktion 2005 (3), MIKROvent GmbH MIKROvent, Mainburg
 http://www.stilsa.com/EN/pdf/optical%20principles%20CSI.pdf, Accessed on 03/10/2016.
 http://ecatalog.stil-sensors.com/STIL-catalog.pdf, Accessed on 03/10/2016.
 J. M. Bennett & L. Mattsson, “Introduction to surface roughness and scattering”, Optical Society of America, Washington DC, 1999.